Perfect for Non-Delivery Oxygen Model
Easy to Use
Compact and Lightweight
Excellent Battery Life
HCPCS Code: E1390, E1392, E1356, E1358, E1399, A4615
PM4150 EasyPulse Portable Oxygen Concentrator (POC)
Height: 10.10 in (25.7 cm)
Width: 6.5 in (16.5 cm)
Depth: 4.5 in (11.4 cm)
Device: 6.6 lbs (2.98 kg)
Shipping: 12.15 lbs (5.5 kg)
AC: 100 - 240 VAC (1.0 A @ 120 VAC)
DC: 11 - 16 VDC (5.0 A @ 12 VDC)
Internal Battery Average Charge Time: 4 Hours
|Oxygen Concentration:||87 to 95% @ 5 l/min setting within 4 to 6 minutes|
Temperature Range: 41°F to 104°F (5°C to 40°C)
Storage Temperature: - 4°F to 140°F (- 20°C to 60°C)
Relative Humidity: Up to 95% Non-condensing
Altitude: Sea Level to 14,000 ft. (4,267 m)
|Outlet Temperature:||Within 2°F (1.1°C) of ambient|
|Noise level:||44 dBA @ setting 2|
12cc ± 2cc
19cc ± 3cc
26cc ± 4cc
33cc ± 5cc
39cc ± 6cc
Peak Flow (l/min)
8.6 ± .3
9.9 ± .2
10.5 ± .3
11.4 ± .2
11.2 ± .2
240 ± 15%
380 ± 15%
520 ± 15%
660 ± 15%
780 ± 15%
|Average Charge Time: 3.5 hrs|
Sensitivity: -0.4 cmH20 @ 20 BPM
Delay Time: ≤0.12 seconds @ 20 BPM
|Cannula:||Maximum 7 ft long standard adult single lumen nasal cannula|
Class II -with respect to protection from electrical shock
Type BF -degree of protection against electrical shock
IP21 -degree of protection against ingress of solids/liquids
|Mode of Operation:||Continuous|
*Specifications are subject to change without prior notice.
The PM4150 Series (EasyPulse POC) meets or exceeds the requirements of the Medical Device Directive(MDD), Council Directive 93/42/EEC Concerning Medical Devices, Directive 2007/47/EC of The European Parliament and Council and all applicable design and safety related standards.
The PM4150 Series (EasyPulse POC) has been FAA Approved. FAA Approved.
LIMITED WARRANTY AND LIMITATION OF LIABILITY
Precision Medical, Inc. warrants that the Portable Oxygen Concentrator POC, (the Product), will be free of defects in workmanship and/or
material for the following period:
Portable Oxygen Concentrator three (3) years from date of shipment
Battery 180 days from date of shipment
Precision Medical, Inc. is NOT responsible for normal wear and tear, or any neglect or abuse of the product. Precision Medical, Inc. is NOT responsible for the shipping cost of the repair. Should any failure to conform to this warranty appear within the applicable period, Precision Medical, Inc. shall, upon written notification thereof and substantiation that the goods have been stored, installed, maintained and operated in accordance with Precision Medical, Inc.’s instructions and standard industry practice, and that no modifications, substitutions, or alterations have been made to the goods, correct such defect by suitable repair or replacement at its own expense.
ORAL STATEMENTS DO NOT CONSTITUTE WARRANTIES.
The representatives of Precision Medical, Inc. or any retailers are not authorized to make oral warranties about the merchandise described
in this contract, and any such statements shall not be relied upon and are not part of the contract for sale. Thus, this writing is a final,
complete and exclusive statement of the terms of that contract.
THIS WARRANTY IS EXCLUSIVE AND IS IN LIEU OF ANY WARRANTY OF MERCHANTABILITY, FITNESS FOR A PARTICULAR
PURPOSE OR OTHER WARRANTY OF QUALITY, WHETHER EXPRESS OR IMPLIED.
Precision Medical, Inc. shall not under any circumstances be liable for special, incidental or consequential damages including but not limited
to lost profits, lost sales, or injury to person or property. Correction of non-conformities as provided above shall constitute fulfillment of
all liabilities of Precision Medical, Inc. whether based on contract, negligence, strict tort or otherwise. Precision Medical, Inc. reserves
the right to discontinue manufacture of any product or change product materials, designs, or specifications without notice.
Precision Medical, Inc. reserves the right to correct clerical or typographical errors without penalty.