Perfect for Non-Delivery Oxygen Model
Easy to Use
Compact and Lightweight
Excellent Battery Life
HCPCS Code: E1390, E1392, E1356, E1358, E1399, A4615
PM4150 EasyPulse Portable Oxygen Concentrator (POC)
Height: 10.1 in (25.7 cm)
Width: 6.5 in (16.5 cm)
Depth: 4.5 in (11.4 cm)
Unit Weight: 6.6 lbs (2.98 kg)
Shipping Weight: 12.15 lbs (5.5 kg)
|Altitude:||Sea level to 14,000 ft (4,267 m)|
|Operating Temperature Range:||41°F to 104°F (5°C to 40°C)|
Temperature Range: -4°F to 104°F (-20°C to 40°C)
Relative Humidity: Up to 95% Non-condensing
|Oxygen Concentration:||87% to 95%|
|Start Up Time||87% O² Concentration ≤ 10 min|
|Oxygen Minute Volume*:||
Setting 1: 240 +/- 15% mL/min
Setting 2: 380 +/- 15% mL/min
Setting 3: 520 +/- 15% mL/min
Setting 4: 660 +/- 15% mL/min
Setting 5: 780 +/- 15% mL/min
|Mode of Delivery:||Pulsed Dose Only|
|Trigger Sensitivity:||0.50 cm H2O Maximum|
|Trigger Delay:||0.11 seconds typical|
AC Power Adapter: 100 - 240 VAC (1.0 A @ 120 VAC)
DC Power Adapter: 11 - 16 VDC (5.0 A @ 12 VDC)
|Battery Duration: (Approximate)||
Setting 1: 4.7 hrs
Setting 2: 3.4 hrs
Setting 3: 2.7 hrs
Setting 4: 2.0 hrs
Setting 5: 1.6 hrs
Times may vary based on atmospheric conditions and battery age.
|Sound (@ Setting 2):||44 dBA|
|Cannula:||Maximum 7 ft (2.1 m) long standard adult single lumen nasal cannula.|
Class II -with respect to protection from electrical shock
Type BF -degree of protection against electrical shock
IPX1 -degree of protection against ingress liquids
Mode of Operation: -continuous
|Volatile Organic Compound (VOC) and Particulate Requirements:
The oxygen delivered from the EasyPulse POC meets the following requirements
for particulate levels, VOC levels, carbon monoxide levels, carbon dioxide levels and ozone levels.
EPA PM: Particulate Matter
ASTM D5466: Ozone Levels
21 CFR 801.415: Carbon Monoxide Levels
EPA NAAQS - Carbon Exposure Limit:s Carbon Dioxide Levels
OSHA Permissible Exposure Limits: Standard Test Methods for Determination of Volatile Organic Chemicals in Atomspheres (Canister Sampling Methodology) *At Sea Level
*Specifications are subject to change without prior notice.
The PM4150 Series (EasyPulse POC) meets or exceeds the requirements of the Medical Device Directive(MDD), Council Directive 93/42/EEC Concerning Medical Devices, Directive 2007/47/EC of The European Parliament and Council and all applicable design and safety related standards.
The PM4150 Series (EasyPulse POC) has been FAA Approved. FAA Approved.
LIMITED WARRANTY AND LIMITATION OF LIABILITY
Precision Medical, Inc. warrants that the Portable Oxygen Concentrator POC, (the Product), will be free of defects in workmanship and/or
material for the following period:
Portable Oxygen Concentrator three (3) years from date of shipment
Battery 180 days from date of shipment
Precision Medical, Inc. is NOT responsible for normal wear and tear, or any neglect or abuse of the product. Precision Medical, Inc. is NOT responsible for the shipping cost of the repair. Should any failure to conform to this warranty appear within the applicable period, Precision Medical, Inc. shall, upon written notification thereof and substantiation that the goods have been stored, installed, maintained and operated in accordance with Precision Medical, Inc.’s instructions and standard industry practice, and that no modifications, substitutions, or alterations have been made to the goods, correct such defect by suitable repair or replacement at its own expense.
ORAL STATEMENTS DO NOT CONSTITUTE WARRANTIES.
The representatives of Precision Medical, Inc. or any retailers are not authorized to make oral warranties about the merchandise described
in this contract, and any such statements shall not be relied upon and are not part of the contract for sale. Thus, this writing is a final,
complete and exclusive statement of the terms of that contract.
THIS WARRANTY IS EXCLUSIVE AND IS IN LIEU OF ANY WARRANTY OF MERCHANTABILITY, FITNESS FOR A PARTICULAR
PURPOSE OR OTHER WARRANTY OF QUALITY, WHETHER EXPRESS OR IMPLIED.
Precision Medical, Inc. shall not under any circumstances be liable for special, incidental or consequential damages including but not limited
to lost profits, lost sales, or injury to person or property. Correction of non-conformities as provided above shall constitute fulfillment of
all liabilities of Precision Medical, Inc. whether based on contract, negligence, strict tort or otherwise. Precision Medical, Inc. reserves
the right to discontinue manufacture of any product or change product materials, designs, or specifications without notice.
Precision Medical, Inc. reserves the right to correct clerical or typographical errors without penalty.